Software and Drivers
Titan D3.1 Meter User Guide
QSIII User Guide
QAM Genacis User Guide
QAM Sniffer User Guide
QAM Sniffer Quick User Guide
QAM Compass User Guide
QAM Shadow User Guide
QAM Marker User Guide
M3 User Guide
Pathfinder Pro II User Guide
The Stick User Guide
SnapTRACK User Guide
SnapTRACK Quick Guide
SnapTRACK Quick Programming Guide
Sentinel User Guide
Sleuth II User Guide
Sniffer Shadow User Guide
Companion User Guide
Fiber Optic Test Equipment Downloads
Sniffer Configuration Utility V1.0.29
Incorporates OUDP settings for Sniffer High Split (S-HS) equipment.
SnifferConfigUtility_V1029.zip file updated February 3, 2023 containing the “SnifferConfigSetup.exe” setup executable. Download zip file, unzip the setup file onto local drive, access “SnifferConfigSetup.exe” file to run installation. 5 firmware files will be extracted into a sub-folder labeled “FW” under the root “SnifferConfig” folder created on the computer’s hard drive. Typical location: “C:\Program Files\ComSonics,Inc\
Setup includes update files;
QAM Shadow DDF firmware version 007
QAM Sniffer DF firmware version 007
QAM Compass firmware version 009
Sniffer High Split (S-HS) firmware version 010
High Split DSP firmware version 002
Sniffer High Split (S-HS) DSP Firmware 005
October 27, 2023
OFDM and OUDP measurement enhancements.
Must have downloaded and installed latest Sniffer Configuration Utility V1.0.29 above.
Download and save the “HS_DSP_V005.zip” file onto your PC. Unzip the and save the “HS_DSP_005.gnx” file. Recommended location: “FW” folder nested under installation default QAM Sniffer Config folder on your “C:” drive ( C:\Program Files(x86)\ComSonics.Inc\QamSnifferConfig\FW).
To update desired Sniffer High Split device: 1) Device must be powered and 2) USB comm cable connected. Access the QAM Sniffer Configuration utility “Firmware Download” option listed under “File” drop menu. In the “FWUpdate” pop up screen, under the DSP Firmware section, select “DSP File” and navigate to the “HS_DSP_005.gnx” file when prompted to select file. Then, select “Upload”. A progress bar will appear along with a completion status notification.
QAM Sniffer III / Sniffer High Split V011 Firmware update April 20, 2023.
Adds automatic bluetooth scan and pair to GQS3/G-HS device by holding QSIII/S-HS bluetooth
button while powering on.
Adds mode to switch from bluetooth GQSIII/ G-HS measurements to local QS3/S-HS measurements
when docked and paired by holding down Bluetooth button until it beeps.
Adds mode to manually switch between bands when device is docked and powered by
holding down the band button down until it beeps. User can then cycle bands by
pressing the band button.
Adds ability to receive configuration from GQS3/G-HS over Bluetooth.
Improved tolerances for OFDM signals which have frequency offsets.
Download and save the “QS3_BINARY_V011” zip file onto your PC. Unzip and save actual “QS3_BINARY_V011” file onto local drive. Recommended location: “FW” folder nested under installation default QAM Sniffer Config folder on your “C:” drive ( C:\Program Files(x86)\ComSonics.Inc\
Utilize the Sniffer Configuration Utility “firmware download” option listed under “File” drop menu to update desired QAM Sniffer DF device. Navigate to the “QS3_BINARY_V011” file when prompted to select file during firmware update process.